Multi-Target Stereotactic Brain Radiation Therapy
MetadataVis full innførsel
- Institutt for fysikk 
Metastasis of the brain is one of the most common and severe neurological complications of cancer. Stereotactic radiation therapy (SRT) characterized by a high dose of radiation, delivered in a single or few fractions, with conformal target coverage and steep dose-falls, has been shown to be effective in treating brain metastases and recurrence gliomas. Historically, Gamma Knife (GK) has been superior to other SRT-techniques.However, the increased flexibility of the new generation of linear accelerators (linac), has made linac-SRTto a competing method. Traditionally, treatment plans for multiple-brain metastases were made with 1isocenter pr.target. Recent studies using volumetric arc therapy (VMAT) showed about the same targetcoverage when optimized with single isocenter. The aim of this study was to investigate the influence of different linear accelerator models and treatmentplanning strategies on irradiated volume of brain normal tissue (V12Gy and V3Gy), conformity (PaddickConformity Index(PCI)) and gradient index (Dose Gradient Index(DGI)). The sensitivity of the locationand number of GTVs was also studied. Treatment plans for different linac-models and treatment planning strategies were made for 4 differentmulti-target brain cases. These plans were supported by the software RayStation (RaySearch LaboratiesAB, v.184.108.40.206 Research). All plans were optimized with the same planning objectives, with some individualcorrections for some cases, until they reached the same acceptable plan criterias Reducing the leaf width of multi-leaf collimator (MLC) reduced the V12Gy, V3Gy and increased the dosegradient (DGI). However, the conformity (PCI) was reduced. Reducing the tip gap, for 5 mm leaf width,showed about the same impact on the volumes treated. It also showed higher PCI compared to the fine-leafplans. The linac at St.Olavs, Elekta Versa HD Agility, was often the least favourable. Single isocenter, especially when combined with new-functionalities in Raystation, showed less sensitivityto location of targets on V12Gy. 1 ioscenter pr.target treated a small volume for distant GTVs, but wereless favourable for close GTVs. The same sensitivity for location was seen PCI. Overall, the PCI for mosttreatment plans was higher than 0.75. For 1 isocenter pr.target on close GTVs PCI was lower than 0.55.The DGI was 0.2-0.3 for most plans, with some single isocenter plans as exceptions. Single isocenter treateda large V3Gy, compared to the other treatment planning strategies for distant GTVs. Based on the result of this project, decreasing the leaf width and leaf tip gap improves the flexibility ofthe linacs. Due to low PCI and less stability for the plans using-fine leaf, the leaf tip gap, rather than theleaf width might be of most importance. Single isocenter might be an attractive and time-effective alternativeto the multiple-isocenter technique, especially when combined with new functionalities in the treatmentplanning system. However, when used for targets located distant to the isocenter, different methods shouldbe tested in hope to reduce the dose spillage. This study was only based on simple cases, the trends observedmust be studied further for more complex cases.